The SoMat eDAQ Bridge Layer (EBRG) offers 16 simultaneously sampled, low-level differential analog inputs through independent connectors. An extremely versatile layer, the Bridge Board works with both amplified and unamplified transducers, including strain gauges, accelerometers, pressure transducers, load cells, and other general analog signals.
The EBRG provides excellent strain gauge conditioning with support for 1/4-, 1/2-, and full-bridges. Automatic balancing and gain settings–as well as software selectable sample rates, excitation, and digital filtering–simplify the set-up of a strain channel. There are several calibration options, including defined value, external, multipoint and shunt calibrations with imbedded software tools. Four shunt calibration resistors per channel are also included, with software selectable shunt direction.
The EBRG is available with an optional Analog Out. Calibration files, directly compatible with popular simulation software, scale the analog outputs to engineering units.
Weight: 4.4lbs (2.0kg)
Temperature: -20° to 65° C
Connectors: (16) SoMat Female Bulkhead ConnectorInputsRange:Minimum full-scale: ±625μV
Maximum full-scale: ±10V
Resolution: 16-bit ADC, across specified full scale range
Sample Rates: Independently Set
100kHZ Master Clock= 0.1Hz to 100kHz
98.3kHz Master Clock= 0.1 Hz to 98,304Hz
Analog: 25kHz, 8-pole Butterworth low-pass filter
Digital: Software selectable; 8-pole Butterworth and Linear Phase
Strain Gage Conditioning:
Supports: 1/4, 1/2, and Full Bridge Devices
Excitation: 5V (Bipolar ±2.5) or 10V (Bipolar ±5)
1/2 and Full Bridge: 100 to 10,000Ω.
1/4 Bridge: Completion Resistor; specified 350Ω (EBRG-350) or 120Ω (EBRG-120)
Four shunt calibration resistors per channel:
50k Ω,100k Ω, 200k Ω, 500k Ω
Selectable shunt direction: upscale (-Sig to -Ex), downscale (-Sig to +Ex)
Ease of Set-up
Configure the eDAQ once, use it in both applications; Analog Output Calibration File directly compatible with popular simulation software
(1) SoMat SAC-TRAN-AO-2 Analog Output Cable
Part Number Code
The following number scheme is used to identify the layer configuration.
EBRG – XXX – XX
eDAQ Layer EBRG
1/4 Bridge Completion Resistor
B = Base
Layer AO = Analog Out